You ask for a repeat EKG to be performed:
The repeat is a completely normal EKG, confirming your "EKG diagnosis": Limb lead reversal, specifically Right Arm-Left Leg Lead Reversal.
To think through how this limb lead reversal causes the changes seen in the original EKG, you have to go back to Eintoven's triangle made up of three bipolar leads ( I, II and III) and three augmented unipolar leads (avR, avL and aVF) (Figure below). The right leg lead serves as the ground. A lead reads as "positive" when the direction of the electrical impulse the same as the direction of the vector. In a normal EKG with a normal axis, the QRS is positive lead I, II and avF because they are directed towards the left side of the heart. aVR is almost always negative because it is directed opposite the cardiac apex.
Lead I Difference between LA and RA electrodes
Lead II Difference between RA and LL electrodes
Lead III Difference between LA and LL electrodes
Lead aVR RA - (LA + LL)/2
Lead aVL LA - (LL + RA)/2
Lead aVF LL - (RA + LA)/2
Right Arm/Left Leg (RA/LL) limb lead reversal is essentially a 180˚ flip of Eintoven's triangle which results in the following changes:
1. aVR and aVL switch places
2. lead I becomes an inverted lead III
lead III becomes an inverted lead I
lead II is an inversion of itself
Take a moment to compare the two EKGs above, and see that this is what has happened.
It can mimic left anterior fascicular block and inferior infarction pattern, but can be suspected based on a positive P, QRS and T wave in lead aVR and extreme right axis deviation.
Limb lead reversals are important to recognize in that they can mimic pathology including myocardial infarction, ectopic atrial rhythms, and conditions characterized by diffuse low voltage (pericardial effusion or pulmonary disease). You want to be able to catch them because you do not want to make critical clinical decisions based on misinformation. There are case reports of changes in patient management that occurred based on incorrectly acquired EKGs that mimicked a pseudoinfarction or junctional rhythm pattern, including thrombolytic therapy in a patient with a baseline abnormal EKG and limb lead reversal [2,3].
Fortunately, other than the sense that "that EKG just don't look right!", there are typical patterns that can clue you in that a limb lead reversal may be present.
General clues that there may be a lead reversal (limb or precordial) include :
1. Abnormal P axis with positive P wave in avR and/or negative P waves in lead I and/or lead II
2. Very low amplitude in the lead I, II, or III
3. Concordant negative QRS and T waves in the leads I, II, III and/or aVF
Abnormal R wave progression in leads V1 to V6, especially when the R wave amplitude increases after an initial decrease in the precedent leads.
The above features can be highlighted by reviewing the typical lead reversal patterns [3,5]. Note images below are all from the same "patient" whose EKGs are featured above.
1. Arm electrode (LA/RA) limb lead reversal: This reversal is the most common limb lead reversal. The tell-tale sign is an inverted p wave, QRS complex, and T wave in lead I. In the absence of dextrocardia, this is pathognomonic of arm electrode reversal. Remember, as general rule I and V6 point the same direction and should look the same on the EKG. aVR and aVF have switched places, so they both look grossly abnormal as well. An example:
|Left Arm - Right Arm Limb lead reversal|
1. Lead I will be a complete inversion of itself
2. Lead II and Lead III will switch places
3. Lead aVL and aVR will switch places
If you notice, both lead reversals involving the right arm lead to an abnormal appearance of lead aVR. One more instance where lead aVR, which is often ignored by emergency physicans and cardiologists alike, can be very useful 
2. Left Arm/Left Leg (LA/LL) limb lead reversal: This is a more subtle limb lead reversal. You can pick up on it by the fact that the p wave is larger in lead I than lead II.
1. Lead I and II switch places (hence the p wave larger in I than II)
2. Lead III becomes an inversion of itself
3. Leads aVL and aVF switch places
3. Right Arm/Right Leg (RA/RL) limb lead reversal: The key to picking up this reversal is an isoelectric recording in lead II.
|Right Arm/Right Leg Limb Lead Reversal|
4. Left Arm/Right Leg (LA/RL) limb lead reversal: The key to picking up this reversal is an isoelectric recording in lead III. The "flat-lining" occurs for the same reason above, except that this time there is no perceived difference between the right arm and the right leg.
|Left Arm/Right Leg Limb lead reversal|
5. Right Leg/Left Leg limb lead reversal usually leads to an EKG that is indistinguishable from normal.
6. Precordial Leads: Alterations in placement of the precordial leads results in abnormal R wave progression pattern and can mimic posterior or anterolateral myocardial infarction.
|Precordial leads altered to the following pattern: V6V5V4V3V1V2|
For more review, Life in the Fast Lane has a great explanation of Eintoven's triangle in limb lead reversals.
Take Home Points: When evaluating EKGs, be on the lookout for the EKG findings suggesting limb lead reversals as these can mimic pathology include acute MI, abnormal atrial rhythms, and conditions causing low voltage. Clues to finding these are abnormal P wave axis, extreme axis deviation, concordant negative QRS and T waves in the leads I, II, III and/or aVF. When in doubt, repeat the EKG.
Submitted by Maia Dorsett (@maiadorsett), PGY-3
Faculty Reviewed by Joan Noelker and Doug Char
1. Peberdy, M. A., & Ornato, J. P. (1993). Recognition of electrocardiographic lead misplacements. The American journal of emergency medicine, 11(4), 403-405.
2. Chanarin, N., Caplin, J., & Peacock, A. (1990). “Pseudo reinfarction”: a consequence of electrocardiogram lead transposition following myocardial infarction. Clinical cardiology, 13(9), 668-669.
3. Guijarro-Morales, A., Gil-Extremera, B., & Maldonado-Martín, A. (1991). ECG diagnostic errors due to improper connection of the right arm and leg cables. International journal of cardiology, 30(2), 233-235.
4. Rudiger, A., Hellermann, J. P., Mukherjee, R., Follath, F., & Turina, J. (2007). Electrocardiographic artifacts due to electrode misplacement and their frequency in different clinical settings. The American journal of emergency medicine, 25(2), 174-178.
5. Harrigan, R. A., Chan, T. C., & Brady, W. J. (2012). Electrocardiographic electrode misplacement, misconnection, and artifact. The Journal of emergency medicine, 43(6), 1038-1044.
6. Pahlm, U. S., Pahlm, O., & Wagner, G. S. (1996). The standard 11-lead ECG: neglect of lead aVR in the classical limb lead display. Journal of electrocardiology, 29, 270-274.